Find the most up-to-date version of ISO at Engineering Intravascular catheters – Sterile and single-use catheters – Part 3: Central venous catheters (ISO ); German version EN ISO. ISO Intravascular catheters — Sterile and single-use catheters — Part 1: General requirements. ISO Intravascular catheters — Sterile and.
|Published (Last):||23 July 2004|
|PDF File Size:||11.9 Mb|
|ePub File Size:||8.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
ISO – Sterile, single-use intravascular catheters — Part 3: Central venous catheters
Round the calculated average water flowrate to the nearest whole number of millilitres. At the time of publication, the editions indicated were valid. Copyright BIS has the copyright of all its publications. This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or grade designations.
The minimum force at break of all other parts of such catheters shall comply with ISO 1: NOTE 4 it is recommended that the distance marks be is cm apart on that portion of the catheter likely to be of im- portance to the isp in positioning the catheter and moni- toring catheter migration. An approved test method for producing a value of 1055-3 will be established.
The following standards contain provisions which, through reference in this text, constitute provisions iiso this part of ISO Table 1 — Minimum force at break of soft tips of lengtli not exceeding 20 mm 4. Search the history of over billion web pages on the Internet.
The details of conditions 105555-3 which the licence for the use of the Standard Mark may io granted to manufacturers or producers may be obtained from the Bureau of Indian Standards. International Standard ISO Attention is particularly drawn to the following: No part of the these publications may be reproduced in any form without the prior permission in writing of BIS. The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
In this adopted standard, reference appears to certain International Standards for which Indian Standards also exist.
BS EN ISO 10555-3:2013
This Indian Standard has been developed from Doc No.: For the purpose of deciding whether a particular requirement of this standard is complied with, the final value, observed or calculated expressing the result of a test 1055-53 analysis, shall be rounded off in accordance with IS 2: An example of a suitable apparatus is shown in fig- ure A. NOTE 3 At the time of publication of this part of ISOthere is no acceptable, validated test method to determine radio-detectability. Until that time, a manufacturer may label his product “radio-opaque” provided he can support this claim by demonstrating that he has an appropriate method for showing radio-opacity.
Review of Indian Standards Amendments are issued to standards as the need arises 10555-3 the basis of comments. Standards are also reviewed periodically; a standard alongwith amendments is iao when such review indicates that no changes are needed; if the review indicates that changes are needed, it is taken up for revision.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the editions indicated: Bureau of Indian Standards BIS is a statutory institution establislied under the Bureau of Indian Standards Act, to promote jiarmonious development of the activities of standardization, marking and quality certification of goods and attending to connected matters in the country.
All standards are subject to revision, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the most recent editions of the standards indicated below.
Disclosure to Promote the Right To Information Whereas the Parliament of India has set out to provide a practical regime of right to information for citizens to secure access to information under the control of public authorities, in order to promote transparency and accountability in the working 1055-3 every public authority, and whereas the attached publication of the Bureau of Indian Standards is of particular interest to the public, particularly disadvantaged communities and those engaged in the pursuit of education and knowledge, the attached public safety standard is made 110555-3 to promote the timely dissemination of this information in an accurate manner to the public.
General requirements IS01 Users of Indian Standards should ascertain that they are in possession of the latest amendments or edition 10555-3 referring to the latest issue of ‘BIS Catalogue’ and ‘Standards: Details of which is given in National Annex A. Certain conventions are, however, not identical to those used in Indian Standards. Technical Corrigendum 1 to 105555-3 above International Standard has been given at the end of this standard. From the first mark, the distance between marks shall not exceed 5 cm.