GOSSEN. GOSSEN. Erlangen. Attachment for. LUNASIX 3. Printed in Germany. e microphotographs. (The following instructions assume that you are. Here are all the pages of the operating manual for the Gossen Lunasix 3 photographic light meter. Lunasix light meters are also labelled Luna-Pro in some parts. You have an older device from GOSSEN? Here you can find the corresponding operating instructions for previous devices in the field of LUNASIX.
|Published (Last):||22 February 2014|
|PDF File Size:||20.94 Mb|
|ePub File Size:||17.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keep this site quiet, please folks, because many of these useful web resources get taken down by spiteful companies that can’t be bothered to put their old manuals online themselves, but will move heaven and earth to stop other people providing that service for free.
The “Profi-System” line continued and ended with the Mastersix.
If you use Pay Pal, use the link below. Fits the ocular tube of all microscopes for exact exposure determination. That’s especially important for close-up measurements. While a selenium element converts light into electric energy, the photo-conductive cell merely changes its conductivity in the presence of light.
After releasing it, you have locked in the correct reading and can take the LUNASIX 3 to better light, or use a flashlight to read the scale.
bossen Index cover page page foldout page page contents, page 1 page page page page page page page page 32 page page page page Micro attachment The Lunasix meter can be used for photo-microscopy, by using the accessory Micro Attachment to fit the meter to a microscope.
Otherwise, fresh batteries must be used.
Gossen Lunasix Instruction Manuals
Measuring comparisions of your LUNASIX 3 with similar or other types luhasix exposure meters cannot be made properly without special laboratory equipment optical bench. The computer dial incorporates standard exposure compensation, exposure factor compensation for filters or close up work, as well as I – IX steps for zone exposure.
Lunasix light meters are also labelled Luna-Pro in some parts of the World. Its microammeter indicates readings in ranges of, ft.
The green and red ring indexes apply only for measurements with the Variable Angle “spot meter” attachment. Flash metering was incorporated, however sensitivity was reduced compared to the Profisix as was the range goxsen accessories Profisix accessories will not fit the Lunasix F.
Gossen Lunasix 3 light meter instruction manual, user manual, free PDF manual
Micro attachment The Lunasix meter can be used for photo-microscopy, by using the accessory Micro Attachment to fit the meter to a microscope. The needle did swing to the left of the scale, but it never can reach to the check line zero line – the very left end line on the scale.
I wondered what wrong with my Lunasix. Ask your dealer for a demonstration! However, you can adapt your LUNASIX 3 exposure readings to the characteristics of your camera, your favorite film, your type of processing, your projector – in other words to your personal touch. The indicator needle should now rest on the check line to the left of the scale divisions When you release the selector, only the scale values applicable to the measuring range which you just selected, remain visible.
Analyze your prints and color transparencies for possible mistakes or errors in judgment. Also ideal for light measurement of the aerial image in photography through terrestrial telescopes. During the battery test, the indicator needle 14 should point to the red check mark Null pointer ambient and flash meter Display: Mallory PX 13 mercury batteries are available at most camera shops.
_MG_3272 Gossen Lunasix F Luna Pro F Manual Handbook the most important features overview
Use a reversal colour film; select several “normal” scenes; take careful reflected and incident lunasixx measurements and shoot!
It’ll make you feel better, won’t it? The ASA setting remains unchanged! With inaccessible subjects, take the incident light measurement at ounasix substitute spot which receives the same illumination as the subject. Therefore, watch the darker areas and expose a little longer rather than too little. The LUNASIX 3 should not be considered a full-fledged footcandle meter because, strictly speaking, footcandles of illumination can be measured only be means of a flat interception screen. Increasing or decreasing the EV number by 1 is equivalent to one stop.
Even I turned the zero screw counter-clockwise, it happened the same way, the needle just swing back and forth, but never reached to the zero line, it reached to 4 of the low light scale, and 13 of the high light scale.
Therefore, keep an eye on such lighter areas and preferably expose a little shorter rather than longer, to get more brilliant colours.
When I kept turning the zero screw clockwise, the needle swinged to the left, then it swing back to the right.