Conmed Sabre ESU Unit | High-quality medical equipment at great prices, unmatched variety, and discounts. () , visit today!. For a period of two years following the date of delivery,. CONMED Corporation warrants the Sabre Electrosurgical Unit against any defects in material or. Find great deals for CONMED Sabre Electrosurgical Unit Shop with confidence on eBay!.
|Country:||Antigua & Barbuda|
|Published (Last):||22 March 2010|
|PDF File Size:||6.66 Mb|
|ePub File Size:||6.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
While wires are conned fully into TB1, tighten 4. Such testing is best carried out by use of an electrosurgical generator tester, as described in Section 4 of the Sabre Service Manual.
Retain orig- inal packing material for future storage or ship- 3. Place spacer static-sensitive devices. This medium-powered generator delivers impressive electrosurgical performance in all modes, including monopolar cut and coag and bipolar coag and blend.
This inspection should The sabee settings option may be changed quite include checks for: Page 32 The A. Since the WDT timing accuracy and ability to may result in damage to other components which interrupt RF output are verified by the micro- had previously been spared by the microproces- processor on every power up, there is no need to sor’s fast response to overcurrents in RUN mode. Got it, continue to print. This section contains information useful in the 4.
You must be logged in to post a review.
Please contact us with all questions and inquiries. While the unit has been Surgical effects are dependent upon a number of designed and manufactured to high industry stan- factors including waveform, electrode size, elec Controls, Displays And Connectors 2. The Sabre should be visually inspected at least every six months. See details for description of any imperfections. The price may be the seller’s own price elsewhere or another seller’s price.
Used CONMED Sabre Electrosurgical Unit For Sale – DOTmed Listing #
To avoid destroying the meter, never activate output terminal connected to Equipment: Take appropriate static washers on the encoder bushings prior to mount- ing on the Control Panel; these washers are thick- control precautions during handling and ser- er than the flat washers used under the encoder vice. This product is already in quote request list. Complies with all current internation- tissue sticking. Food and Drug Administration and state and local regulatory agencies.
This permits a more thorough eval- provide output, but also forcing the WDT to lock uation of lamp integrity than is possible during up permanently if WDT-Q2 ever fails. The power delivered to the output at mains voltage zero-crossing is nearly zero.
This item doesn’t belong on this page. Comments to this Manuals Your Name.
Advertise Increase Your Brand Awareness. Referring to Figure 5. Same as in Section 4. Power Amplifier Waveforms Figure 3. Page 26 Usually accompanied by audible or visible sparking. This results in output current dental meter failure.
Show More Show Less. Only the activated accessory will be live.
Conmed Sabre 180
Restoring a point to Calibrated status requires 3. Installation And Operation Conmed representative immediately. If so, do not bid on this item unless you are an authorized purchaser. This section contains information for ordinary upkeep of the Sabre If all four fail to cause the unit to Section 4. Tone Generator lines, WV0-WV7, determine which waveform is that is switched sbre and off at the frequency of the selected by the microprocessor, and the lower desired tone.
Verify security of the connection by pulling on the 18 with The factory configures the unit to power up using the pliers. Page 35 will draw collector current in proportion to its 18 PA consists of a single power MOSFET dri- base current, which in turn is controlled by ving the emitters of three bipolar transistors.
When the Top Refer to Figure 5. Page 6 This page intentionally left blank. Usually accompanied by audible or visible sparking. The sale of this item may be subject to regulation by the U.
Continuity Detector Troubleshooting Check the Base Voltage Generator and Waveform The open circuit voltage on the switching lines Generator for proper function before proceeding. Page 9 purged and filled with non-flammable gas prior possible from the electrosurgical site and the to abdominal surgery.
December 27, In Stock: Fast blow V fuses rated at the swbre full 4.
A11, and A12 cause the corresponding output Y0 -Y7 to go low. Provides unsurpassed electrosurgical effects in xabre coagulation mode. OEM Medical Solutions view seller’s other items: This diagram illustrates the functional parti- This section is located on the forward end of the tioning of the unit. The lowest-priced item that has been used or worn previously.