First middle-atmospheric zonal wind profile measurements with a new ground- based microwave Doppler-spectro-radiometer. R Rüfenacht, N Kämpfer, A Murk.  S. C. Müller, A. Murk, C. Monstein, N. Kämpfer, and H. Meyer, the equal filter characteristics of X, MONTH YEAR Axel Murk was born in Germany in Karl Jacob, Arne Schröder, and Axel Murk. Abstract—This ; axel. [email protected]). Axel Murk was born in Bad Tölz, Germany, in
|Published (Last):||17 June 2017|
|PDF File Size:||14.81 Mb|
|ePub File Size:||8.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Dr. Axel Murk
Additionally the channels of an AOS radiation from the atmosphere from all spectrometer types were in good agreement. The frequency axis is ENOB 6. The spectral resolution around 0 20 40 60 80 line center is 61 kHz. There are two possibilites to compare the retrieval bank with 44 channels and filter bandwidth between kHz characteristics of the different spectrometers.
The main reason for V. As long as there are no drift is directly the power spectrum. He is currently working on a broad band solar polarization radio spectrometer. Consequentially we had to filter the time. The cost can be reduced by a factor ofnot Transactions on Geoscience and Remote Sensing, vol.
Axel Murk – Google Scholar Citations
It is best to turn the mirror as few as possible. His research interests are microwave remote sensing of the atmosphere and optics in the millimeter and submillimeter range. In general the agreement between  A. We assume a linear behaviour of the instrument between hotload, coldload Fig. The spectral line of such transitions is pressure is needed to know for the calibration of the instrument before broadened, which allows one to retrieve an altitude profile performing real atmospheric measurements first from ground of the species under investigation.
AOS and a modelled line.
Dynamical range and linearity in power Signal IN Spectrum OUT In the calibration we assume a linear relationship between measured voltages of the hotload, coldload and the atmo- Fig. When dealing with broadband noise sources the performance normalized amplitude 0. To avoid stepping problems the power was measured in parallel with a HPA power meter, too. The identical Allan times. But this can only be asel with continous-wave signals.
Rodgers, Inverse Methods for Atmospheric Sounding: Then we measured  C. TB of the atmosphere is determined by LO 3. This is reflected in a higher upper with heterodyne receivers for microwave remote sensing of altitude level of the broadband Axxel spectrometer than of trace gases of the atmosphere.
Help Center Find new research papers in: The difference is clearly longer Allan time than the Wxel. For this 50 MHz bandwidth and 75 kHz resolution was used for the reason the novel approach of using real-time digital Fast Fourier observation at line center. An FFT spectrometer has small dimensions, low weight, and lower costs when compared to sxel Observations of critical constituents are important in order to AOS with an equal bandwidth and equal number of channels.
There he was responsible for all kind of high speed broadband radio spectrometers. Up to now we from the IAP perform ground-based and airborne measurements.
European Planetary Science Congress, Thus Then, during two flight campaigns in November and recalibration of the frequency axis is needed and hence the Octoberparallel water vapor measurements from the loss of one measurement. The wass downconversion problem but this would only shift the problem was operated in total power mode and was calibrated using towards aliasing effects at both edges of the filtered band.
Sensors 10 wxs, The first sidelobe level is at 13 dB. IEEE transactions on microwave theory and techniques 58 7, The measured brightness temperature onboard. Thus system tests have been by the number of samples taken for one calculation. A turning mirror switches between the three positions. Difficulties in downconversion to the first Nyquist band. The total integration time is AOS linecenter 4. Verified email at iap.
He was then chief designer of automatic in-process test systems at Alcatel company borj Switzerland. He received the M. Email address for updates. The sensitivity has effects on the non-linear channel axis 0 as demonstrated in Figure 8.
Remember me on this computer. Murk, and by a factor of 10 compared to the AOS.
FFT spectrometer channels are convolved the first hour of the flight. If the LO and the atmospheric measurement. The effective bandwidth used for data processing in mixer to convert the signal down to an intermediate frequency the experiment was MHz for respectively MHz of a 3.
We measured the power of a broadband noise source with the spectrometers. This means that if we decrease the 1 temporal resolution we get a higher bor resolution. European Planetary Science Congress 8 As an example before constituents using this novel approach with digital FFT spectrom- eters and its comparisons to measurements from the AOS and take-off ambient air enters the cabin and can cause temperature filterbank.