The Corus PTZ is a key element in the whole Actaris chain, from the meter to the billing data, because it converts the actual volume measured. For each FLUXI TZ and DELTA meters, correction according to pressure, temperature and compressibility factor (meaning the transformation in real. CORUS is an electronic volume converter dedicated to commercial and industrial applications. It converts the actual volume measured by the gas meter to.
|Published (Last):||25 August 2005|
|PDF File Size:||11.13 Mb|
|ePub File Size:||20.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
It must be preferentially screwed to the “Pr” reference pressure tap of the meter: The tamper input can be set into one of the following modes: Its volume input is a low frequency one that can be connected to any Ex compliant gas meter that provides such information.
Solenvis SL79 Itron-Corus gas volume converter
When such a alarm occurs: The Corus Gas Volume Converter is an electronic volume converter dedicated to commercial and industrial applications. All daily consumptions are stamped from the gas hour to the gas hour of the next day.
The interval can be selected among: Input pulse weight, unconverted Index, Converted Index: The terminal J6 that correspond to the serial port is located in the customer area: Period 1 second Retrans. The respect of this directive is mandatory from July,1st and is now included in the ‘CE’ marking.
Switchthe following procedure has to be followed: Place the metrologic cover and seal it?
The connection of the CORUS to any other instrument or device must be done in accordance to the electrical parameter indicated in the certificate. The generally accepted installation actais are: The five keys allows to move into the various menus as described below. Thanks to its accurate piezo-resistive pressure sensor and its 4 wires PT temperature probe, CORUS provides an accurate conversion on the whole temperature range. The activation duration of the alarm outputs can be selected.
Once the configuration is completed and your data points are registered and subscribed to Wattics, you will receive access to your Wattics dashboard, and from then on will be able to log in to your Actxris Dashboard at http: Installation and program procedures must be carried out and inspected by qualified personnel.
The pressure sensor intends two cogus holes allowing to seal and link the sensor with the meter installation. Are you ready for energy savings?
It can withstand an overpressure of 1. This input shares the same terminals J3 and J4 and the same cable as the main LF corsu. The corresponding marking is: CORUS uses the measured working values of volume, pressure and temperature to provide: It allows to check the good operating of the metering function and of the pulse emitter of the meter.
Corus PTZ manages three temperature alarms: Corus PTZ manages three pressure alarms: In case of failure, the magnetic detection can be overridden coris turning on the user interface. It operates at baud, baud, baud, baud, baud, baud.
When changing the battery, the battery counter has to be re-programmed so as to remove the battery alarm.
Energy Learning community online Be part of the big picture, share your knowledge and get access to actaeis and expertise from others Visit our forum. It operates at baud, baud, baud, baud. LF input pulse weight?
Proofreader of volume of gas SEVc-D (Corus) PTZ, TZ, Itron (Actaris).
A hinge on the left of the box makes any operation on the field very easy, providing a clear access to all major parts of the product terminals, switches, internal battery,… Corus PTZ is built with three electronic boards: The non-metrological firmware can be updated in the field without stopping operations and without breaking the metrological seals MID. They can be either: It is designed to calculate gas volumes under base conditions, based on a volume at metering conditions, a temperature, a pressure, a compressibility.
RAM and flash memories, …? When a pressure alarm occurs: You may have a signal barrier in between to separate the safe zone from the hazardous zone.
The fixed part includes the following data: For each event, following data are available: During the main cut, the internal backup battery will ensure the total functioning of Corus PTZ and the battery counter will be decreased with the same management described in the “battery alarm” chapter. The alarm outputs can remain active: Based on the use of flash memory, its architecture allows the download of a new firmware release through a laptop without board pfz.
EnergoBit / Gas Volume Converter CORUS (PTZ)
All alarms will be reset. The frequency of retransmissions indicated in the table correspond to average frequency on the 2 outputs, when both channels are configured for pulses. All pressure sensors are delivered with their own characterization coefficients 12 that must be programmed into the converter so that the accuracy required by EN can be reached over the whole pressure and cataris range.